LH Ovulation Rapid Test Kit
|Name||LH Ovulation Rapid Test Kit|
|Features||High sensitivity, Simple, Easy and Accurate|
|Specification||3.0mm 4.0mm 5.5mm 6.0mm|
|Shipping||By sea/By air/TNT/Fedx/DHL|
|Instrument classification||Class II|
|Certificate||CE/ ISO13485 /FDA|
|Shelf life||two years|
|Type||Pathological Analysis Equipments|
Principle of FLH Rapid Test Device
The test reagent is exposed to urine, allowing urine to migrate through the absorbent test strip. The labeled antibody-dye conjugate binds to the LH in the specimen forming an antibody-antigen complex. This complex binds to the anti-LH antibody in the test region (T) and produces a color line. In the absence of LH, there is no color line in the test region (T). The reaction mixture continues flowing through the absorbent device past the test region (T) and control region (C). Unbound conjugate binds to the reagents in the control region (C), producing a color line, demonstrating that the test strip is functioning correctly. The test strip can accurately detect your LH surge when the concentration of LH is equal to or greater than 25mIU/ml.
Read the entire procedure carefully before performing any tests.
Allow test strip and urine specimen to equilibrate to room temperature (20-30℃ or 68-86℉) prior to testing.
1.Remove the test strip from the sealed pouch.
2.Holding the strip vertically, carefully dip it into the specimen with the arrow end pointing towards the urine.
NOTE: Do not immerse the strip past the Max Line.
3.Remove the strip after 10 seconds and lay the strip flat on a clean, dry, non-absorbent surface, and then begin timing.
4.Wait for colored lines to appear. Interpret the test results at 3-5 minutes.
NOTE: Do not read results after 10 minutes.
CONTENTS, STORAGE AND STABILITY
The test strip consists of colloidal gold-monoclonal antibody against LH coated on polyester membrane, and monoclonal antibody against LH and goat-anti-mouse IgG coated on cellulose nitrate membrane.
Each pouch contains one test strip and one desiccant. Each box contains five pouches and one instruction for use
INTERPRETATION OF RESULTS
If the test line is equal to or darker than the control line, you are experiencing the hormone surge that indicates you will ovulate soon, usually within 24 to 48 hours of the surge. If you want to be pregnant, the best time to have intercourse is after 24 hours but before 48 hours.
Only one color line appears in the control region, or the test line appears but is lighter than the control line. This means there is no LH surge.
The result is invalid if no color line appears in the control region (C), even if a line appears in the test region (T). In any event, repeat the test. If the problem persists, discontinue using the lot immediately and contact your local distributor.
NOTE: A color line appearing in the control region can be seen as a basis for effective testing.
We, Hangzhou Testsea Biotechnology Co., Ltd is a fast-growing professional biotechnology company specialized in researching, developing, manufacturing and distributing of advanced in-vitro diagnostic(IVD) test kits and medical instruments.
Our facility is GMP, ISO9001, and ISO13458 certified and we have CE FDA approval. Now we are looking forward to cooperating with more overseas companies for mutual development.
We produce fertility test, infectious diseases tests, drugs abuse tests, cardiac marker tests, tumor marker tests, food and safety tests and animal disease tests, in addition, our brand TESTSEALABS have been well known in both domestic and overseas markets. Best quality and favorable prices enable us to take over 50% the domestic shares.
6.Pack the pouches
7.Seal the pouches
8.Pack the box